The VR-720SS Flushing Tool is developed to enable staff to clean the ophthalmic instruments
either manually or by other means, prior to sterilization.
A vitreoretinal instrument is relatively delicate and it is vital that care is taken when
cleaning and sterilizing them.
Also for use on Micro scissors and Micro Capsulorhexis forceps from the Anterior Range.
OPHTHALMIC INSTRUMENTS - CLEANING SYSTEM
VR-720SS Flushing Tool
Instructions for use:
Remove cap from flushing Instrument.
Fit Delrin cap to Nose of Flushing Instrument.
Close the handle & fit to the Delrin cap and nose cone of the Flushing Instrument.
Push in firmly until the handle comes to a stop.
Release the arm of the handle so that it becomes fully open.
This allows the flushing medium to pass through the instrument.
Replace the barrel of the Flushing Instrument and ensure it is tightly closed.
For Manual Cleaning fit the syringe with the flushing medium to the Luer Lock of the
Flushing Equipment securely and flush instrument.
Apply continuous pressure to the syringe until water exits the tip.
For Automated Dishwasher cleaning fit the Flushing Equipment directly to
the Luer Lock in the dishwasher.
Run cycle in accordance with recommended instructions.
Instructions for use